<!--
 * @Author: cgl
 * @Date: 2023-04-18 15:12:26
 * @LastEditors: cgl
 * @LastEditTime: 2023-09-04 11:32:31
 * @FilePath: \public-health-emergencies-fed:\新放射\新radiation-hygiene-fe\src\views\person\itemControl\item10.vue
 * @Description: 
 * 
 * Copyright (c) 2023 by cgl/微盛信息产业有限公司, All Rights Reserved. 
-->
<template>
	<table border="1" cellspacing="0" cellpadding="0" align="center">
		<tr class="title">
			<td rowspan="2">序号</td>
			<td rowspan="2">检测项目</td>
			<td rowspan="2" width="20%">检测方法及条件</td>
			<td rowspan="2">验收检测判定标准</td>
			<td rowspan="2">状态检测判定标准</td>
			<td rowspan="1" colspan="2">稳定性检测</td>
		</tr>

		<tr class="title">
			<td>判定标准</td>
			<td>周期</td>
		</tr>

		<tr>
			<td >1</td>
			<td >标准照片密度</td>
			<td >4cmPMMA</td>
			<td >1.4OD～1.8OD；<br>建立基线值</td>
			<td >基线值±0.2OD内</td>
			<td >基线值±0.2OD内</td>
			<td >一个月</td>
		</tr>
		<tr>
			<td >2</td>
			<td >AEC响应</td>
			<td >2cm、6cm厚模体与4cm厚模体影像光密度差</td>
			<td >与4cm的值相比<br>±0.2OD内</td>
			<td >与4cm的值相比<br>±0.2OD内</td>
			<td >与4cm的值相比<br>±0.2OD内</td>
			<td >六个月</td>
		</tr>
		<tr>
			<td >3</td>
			<td >高对比度分辨力</td>
			<td >高对比度分辨力测试卡，分别0°和90°放置</td>
			<td >≥10.0lp/mm</td>
			<td >≥10.0lp/mm</td>
			<td >——</td>
			<td >——</td>
		</tr>

	</table>
</template>

<script>
export default {
  name: "item40",
  data() {
    return {
    };
  },
  created() {
  },
  methods: {
  }
};
</script>

<style lang="scss" scoped>
@import './content.scss';
</style>